TWP/SMET/C1 - Reprocess: Skewed wind direction 1 data
|DataStreams:||twpsmet60sC1.a1, twpsmet60sC1.b1, twpsmet60sC1.00|
Wind direction 1 data is incorrect. Manus SMET wind direction for Wind-1 (upper unit) was apparently skewed from its original position as a result of checking direction calibration. As near as we can tell the position of the sensor was rotated clockwise Somewhere in the range of 45-90 degrees, most likely in steps 22.5 degrees. Data comparison performed on 3-2-00, using a small amount of data from this instrument.
Do not use data without careful analysis and correction.