SGP/EBBR/E9 - Right Aspirator Not Working
|DataStreams:||sgp5ebbrE9.b1, sgp30ebbrE9.b1, sgp15ebbrE9.b1|
The result of the right aspirator not working was a reduction in measured temperature gradient and thus a reduction in estimated sensible heat flux and an increase in latent heat flux. The fluxes could be in error by as much as 10%. Reported right air and humidity probe temperatures, top and bottom air and humidity probe temperatures, right relative humidity, and top and bottom vapor pressure are affected.