SGP/THWAPS - RH low by factor of 1.1
|DataStreams:||sgpthwapsB1.b1, sgpthwapsB1.a1, sgpthwapsB1.00|
Because of instrument mentor stupidity, all B1 TWHAPS RH values collected since the specified being date are low by a factor of 1.1. This problem also will affect the vapor pressure values in the file, but the scaling will not be linear. See companion DQR for THWAPS C1 for more detail.
The CSI data logger needs to be reprogrammed. Data users can multiply the values in the file by 1.1 to obtain the correct RH.