SGP EBBR E13 - E13 EBBR Left Air Temperature Incorrect
|DataStreams:||sgp30ebbrE13.b1, sgp5ebbrE13.b1, sgp15ebbrE13.b1|
This condition occurred most of the time during the indicated period (hence color yellow). The left air temperature was incorrect only when the AEM was in the home 30 position. This only affected the bottom air temperature calculation. However, it made the latent heat flux, sensible heat flux, and Bowen ratio incorrect. The sensor was improperly inserted into the aspirator shield and was pinched between the shield and the T/RH probe. When the AEM was in the home30 position, the aspirator hose position may have caused the sensor to short to ground.
Correct heat fluxes cannot be recalculated for this period. Flag the data with this comment.