SGP/EBBR/E20 - Pressure Sensor Data Incorrect
|DataStreams:||sgp5ebbrE20.b1, sgp30ebbrE20.b1, sgp15ebbrE20.b1|
The E20 pressure sensor suffered from spikes in pressure, which were related to increases in voltage during the day as the solar panels charged the battery. These spikes were not of a magnitude that would affect the 30 minute surface fluxes. Replacement of the pressure sensor solved the problem.