TWP/MWR/C1 - wrong azimuth
The MWR was initially installed at an azimuth angle defined as 180 degrees but the value in the configuration file was not changed from the default of 0 degrees. In examining photos taken during the installation of the AWS tower, I noticed that the MWR was rotated opposite the normal orientation. The value in the configuration file was changed to reflect the actual azimuth of the instrument.
The scientifically relevant data are not affected.