NSA/MPL/C1 - Instrument problem
|DataStreams:||nsamplC1.a1, nsamplC1.00, nsamplpsC1.a0|
On 2/18, the energy monitor on the MPL dropped from 11 microjoules per second to 2 microjoules per second. This severely affected data quality. This appears to be have been an intermittent cable problem for the laser Q-Switch. A new cable was provided and installed on 2-24.
No work around is available.