NSA/METTWR/C1 - CMH Dew Point and RH data Incorrect
The Chilled Mirror Dew Point reported negative dew point values. This caused the RH to be much lower than reality also. Compared to the 2m level the RH and dew point from the CMH are incorrect. Mirror was dirty, it was removed, cleaned and recalibrated.