TWP/SKYRAD/C2 - UVB Data questionable/incorrect due to uncontrolled sensor temperature
The UVB was not temperature controlled to 25C as required to ensure accurate data. UVB data should ONLY be used after checking both the bit_data and twpskyrad20sC1.a1 uvb_temp or inst_down_uvb_temp. UVB data where the UVB temperature was outside the range of 24-26C should not be used.
Use all UVB data from Nauru with caution, checking the uvb_temp info in the twpskyrad20sC2.a1 data stream.