NIM/MWRP/M1 - Reprocessed: IRT configuration setting error
The infrared thermometer settings (IRT) were not configured as specified by the manufacturer. The analog output was set to a range of -60 to +50 C (0-10V) instead of -50 to +50 C (0-10V). The settings were changed on April 20. The IR temperatures should be slightly smaller than reported. The data have been reprocessed to correct the IRT calibrations. Reprocessed data were archived in August 2006.
Retrieve the reprocessed data for use in your analysis.