|Start Date||Start Time||End Date||End Time||Data Quality Metric|
TWP/MFRSR/C1 - Installation Issue
|DataStreams:||twpmfrsrC1.a0, twpmfrsrlangleyC1.c1, twpmfrsrlangplotC1.c1, twpmfrsraod1michC1.c1,|
A refurbished system with a new logger was installed on 20100219. This was our first install in the southern hemisphere. We had an unknown software bug, and initially the instrument was set up 180 deg off in an effort to get it to shade properly. This didn't work, and I suspect is related to the weird neg dir normal values that started this DQPR. A software fix was made, and the instrument was set up correctly on 20100512.