TWP/SMET/C1 - Missing RH, Barometer, and ORG data
Atmos_pressure data suddenly went missing on 03/02 at 07:34 UTC and consequently, the relative humidity data flatlined. Furthermore, after a sudden precipitation event when the max precip rate flag was tripped the TBRG total has not agreed with the comparison ORG or the precip rate plot. A loose wire connection was fixed 3/22 restoring RH and barometer data. A damaged cable was repaired 5/11 restoring ORG data.