SGP/EBBR/E20 - Incorrect Sensible and Latent Heat Fluxes
|DataStreams:||sgp5ebbrE20.b1, sgp30ebbrE20.b1, sgp15ebbrE20.b1|
Because of a periodic AEM failure, home signal 30 was often stuck at the home signal 15 position. These occurrences resulted in incorrect latent and sensible heat flux values at times. Some missing data occurred because of low battery voltage (the CR10 program stops running when battery voltage is insufficient. The data outage events were preceded and followed by several hours of bad/suspicious data.