GRW/MWR/M1 - Erratic brightness temperature
Around 12 Z on 7/20 both the PWV and 23.8 GHz brightness temp values began to diverge from those of the MWRP. By 7/22 both the LWP and 31.4 GHz brightness temp had followed suit. At 0600 Z on 7/22 multiple (10-20), brief (<=1 min) data outages began appearing hourly. The problem persisted intermittently until 8/9. On 8/9, changes in the settings of the computer restored the proper functionality of the MWR software.
Use MWRP during specified period.