GRW/WACR/M1 - Reprocessed: Reflectivity and Height Offsets
|DataStreams:||grwwacrspeccmaskcopolM1.a0, grwwacrM1.b1, grwwacrspeccmaskxpolM1.a0|
The instrument vendor informed us of a range (height) offset of 13 m and provided updated reflectivity calibration. All values reported for 'Reflectivity' were approximately 3 dB too low. In addition, the Reflectivity in the first and fourth range gates (starting from the surface) needed to be further corrected by an additional 2.7 and -0.9 dB, respectively. The 'height' field values were 13.33 m too low. The data were reprocessed to correct these problems. The grwwacrM1.b1 data were reprocessed and archived in October 2010. The grwwacrspeccmaskcopolM1.a0 and grwwacrspeccmaskxpolM1.a0 data were reprocessed and archived in January 2012.
Retrieve reprocessed data.