NSA/GVRP/C1 - Incorrect surface temp and RH
The instrument experienced a period of missing data on 3/23/2011 from 05:38:30 GMT until 08:21:44 GMT. When data returned the surface temperature and surface relative humidity were incorrect. Surface temperature flat-lined at 999.990 K and surface RH flat-lined at 999.990%. On 3/24 the sensor was removed and cleaned resolving the problem.