SGP/SWATS/E13 - Failed sensor W 25cm
After a lightning strike near the CF, the SWATS trise values were reporting near zero. The current channel in the CE8 module failed, so the wire was moved to the next channel and trise values returned to normal, except for the West 25cm sensor. The trise for this sensor remained at zero, and did not change after re-seating the connections. Likely sensor has failed.
Use the East 25cm sensor.