MAO/AOS/M1 - Weak solder connection in dilution flow meter comport cable
|DataStreams:||maoaosccn100M1.a1, maoaosclap3wM1.a1, maoaoscpcM1.a1|
The dilution flow meter reading was erratic. Plots of the data suggest that the flow varied by about 5%. Problem was corrected when wire was re-soldered.
For more precise data use the b1 level data as correction has been applied to this data. For a good approximation multiply concentrations, scattering and absorption coefficients by 1.35 to obtain approximate dilution correction.