|Start Date||Start Time||End Date||End Time||Data Quality Metric|
|07/18/2015||0000||07/20/2015||2359||Does not affect quality|
SGP/AERI/C1 - Instrument run in slow sample mode.
|DataStreams:||sgpaerich2C1.a2, sgpaeri01ch2C1.a1, sgpaeriC1.00, sgpaeri01C1.00, sgpaerixC1.00,|
sgpaerisummaryC1.a1, sgpaeriengineerC1.b1, sgpaerisummaryC1.a2, sgpaeri01engineerC1.a1,
sgpaerich1C1.b1, sgpaeri01summaryC1.a1, sgpaerisummaryC1.b1, sgpaerich1C1.a1,
sgpaeri01ch1C1.a1, sgpaerich2C1.b1, sgpaerich2C1.a1, sgpaerich1C1.a2
Data is fine but the instrument was run in slow sample mode for these 3 days. We had changed the digital meter followed by a short 3rd body test that is run in slow sample mode. The instrument wasn't not switched back to rapid sample mode during this period.
Be aware that rather then records every 20 seconds there are records approximately every 8 minutes.