|Start Date||Start Time||End Date||End Time||Data Quality Metric|
ENA/MFRSR/C1 - Heater not functioning - bad connection
The heater was not functioning during the period covered by this DQR. After trying different sensors and system components we determined the problem was with a connector. Re-seating the cable resolved the problem.