OLI/AERI/M1 - Instrument
|DataStreams:||oliaerich2M1.b1, oliaeriengineerM1.b1, oliaerich1M1.b1, oliaerisummaryM1.b1|
During this period the stirling cooler was failing. Channel 1 data collected when the detector was above 80 K is suspect and channel 2 is likely useable.
Check the detector temperature when picking out data to use.