OLI/DL/M1 - Scanner stuck at 0 degrees elevation
|DataStreams:||olidlusrM1.b1, olidlppiM1.b1, olidlcal2M1.b1, olidlfptM1.b1|
The elevation scanner motor was not working properly due to a software glitch. The scanner was stuck at an elevation angle of 0 degrees. The azimuth scanner continued to work properly, and all reported beam angles are correct (albeit 0 for the elevation angle). As a result, there is no vertical staring data during the period covered by this DQR. Also, the PPI scan data can not be used to compute vertical profiles of horizontal winds. This malfunction had no impact on the quality of the radial velocity, backscatter, or intensity data.