|Start Date||Start Time||End Date||End Time||Data Quality Metric|
MAO/AERI/M1 - White light failure
|DataStreams:||maoaerisummaryM1.b1, maoaeriengineerM1.b1, maoaerich2M1.b1, maoaerich1M1.b1|
No data during this period. White light had failed. Instrument repairs onsite by Jonathan Gero were successfully completed and instrument was back operational on December 1. End of deployment calibration verification tests passed, instrument was packed and is ready to be installed at the next site.