Subject:  SGP/BRS/C1  Reprocess: Incorrect calibration coefficients 
DataStreams:  sgpbrs20sC1.a0, sgpbrsC1.b1

Description:  An incorrect serial number and calibration coefficients were used in processing.
Diffuse:
SN 33785 is correct. CF 107.57 is wrong; should be 107.45
(8/14/15 19:24 Sensor swap: 33785 removed. 33238 installed)
Perform the following calculation to correct the data:
DD= down_short_diffuse_hemisp:
Instrument: SN 33785
Old (incorrect) CF: 107.57 [W/m2\mv]
New (correct) CF: 107.45 [W/m2\mv]
DD (new) [W/m2] = DD (old) [W/m2] * 107.45 [W/m2\mv] / 107.57 [W/m2\mv]
Global:
SN 30811 is correct. CF 120.63 is wrong; should be 120.77
(8/14/15 19:24 Sensor swap: 30811 removed. 30812 installed)
Perform the following calculation to correct the data:
DS= down_short_hemisp:
Instrument: SN 30811
Old (incorrect) CF: 120.63 [W/m2\mv]
New (correct) CF: 120.77 [W/m2\mv]
DS (new) [W/m2] = DS (old) [W/m2] * 120.77 [W/m2\mv] / 120.63 [W/m2\mv]
NIP:
SN 34504 is correct. CF 123.18 is wrong; should be 123.19
(8/14/15 19:24 Sensor swap: 34504 removed, 29851 installed)
Perform the following calculation to correct the data:
NIP= short_direct_normal
Instrument: SN 34504
Old (incorrect) CF: 123.18 [W/m2\mv]
New (correct) CF: 123.19 [W/m2\mv]
NIP (new) [W/m2] = NIP (old) [W/m2] * 123.19 [W/m2\mv] / 123.18 [W/m2\mv] 
Suggestions:  Wait for reprocessed data. 
Measurements:  sgpbrs20sC1.a0:  inst_global
 inst_direct_normal
 inst_diffuse
more
sgpbrsC1.b1:  short_direct_normal_min
 down_short_hemisp_min
 down_short_hemisp
 down_short_diffuse_hemisp_min
 short_direct_normal
 down_short_diffuse_hemisp
 short_direct_normal_std
 short_direct_normal_max
 down_short_hemisp_max
 down_short_hemisp_std
 down_short_diffuse_hemisp_std
 down_short_diffuse_hemisp_max
more
