|DQR ID||Subject||Data Streams Affected|
|D050614.1||SGP/VCEIL/C1 - poor signal||sgpvceil25kC1.b1|
|Start Date||Start Time||End Date||End Time|
SGP/VCEIL/C1 - poor signal
When parameters were changed (set noise_h2_compensation to "ON" and units to "METERS") to correct a low amplitude problem, the backscatter plots began displaying horizontal stripes indicating detector "ringing". However, the cloud base heights are not affected. The problem was corrected when the instrument was replaced.