|DQR ID||Subject||Data Streams Affected|
|D030515.3||NSA/MWR/C1 - no air temperature signal||nsamwrC1.00, nsamwrlosC1.b1, nsamwrtipC1.a1|
|Start Date||Start Time||End Date||End Time|
NSA/MWR/C1 - no air temperature signal
|DataStreams:||nsamwrC1.00, nsamwrlosC1.b1, nsamwrtipC1.a1|
When the new blower was upgraded by Radiometrics and reinstalled on the MWR, the air temperature sensor failed to properly report. It was determined that the wires carrying the signal to the analog board did not conform to the standard expected by the upgraded blower. The problem was corrected by changing the wiring and modifying the MWR software to read the signal from the appropriate corresponding channel.