|DQR ID||Subject||Data Streams Affected|
|D050718.2||PYE/ECOR/M1 - Reprocess: Sensor aligned to Magnetic North||pye30ecorM1.b1|
|Start Date||Start Time||End Date||End Time|
PYE/ECOR/M1 - Reprocess: Sensor aligned to Magnetic North
The SMET wind direction was consistently 15 to 20 degrees larger than the wind direction determined by the ECOR. The ECOR was mistakenly aligned to magnetic north during this period. The magnetic declination is 14 deg. 58 min., which explains most of the difference.